Defibrillator Waveform: Biphasic Charge Time: Less than 7 seconds with a new fully charged battery. Energy Display: Monitor display indicates both selected and delivered energy. Charge Prompt Type: Voice and visual prompts. Electrode Impedance Measurement Range: 0-250 ohms.
ECG Monitoring Patient Connection: 3-lead ECG cable, or 5-lead ECG cable, paddles. Lead Selection: Displayed on monitor, paddles, I, II, III, AVR, AVL, AVF, V. ECG Size: 0.25, 0.5, 1, 1.5, 2, 4 cm/mV diplay on monitor. Heart Rate: 20-300BPM. Heart Rate Alarm: On/Off displayed on monitor, user-selectable. Smart Alarms: Beeper/voice prompts indicate shockable rhythm.
Battary Type: Rechargeable, Ni-MH battery, 12V. Operating Time: For a new, fully charged battery: 60 defibrillator discharges at maximum energy, or 3 hours minimum of continuous ECG monitoring. Additional parameters will effect operating time with different functions.
Manual Mode Energy Selection: Selectable at 2, 5, 7, 10, 20, 30, 50, 70, 100, 150, 200, 300, 360 joules. Synchronized Mode: Synchronizes defibrillator pulse to patient's R-wave. "SYNC" message displayed on monitor.