Qinhuangdao, Hebei, China
Member Since: 2008
Business Type: Manufacturer/Factory & Trading Company
|NIBP monitoring||Measurement method: Oscillometry|
Measurement mode: manual/auto.
Auto measurement interval: adjustable from 1 to 255 minutes.
Accuracy: Maximum Mean deviation±5mmHg, Maximum Standard deviation≤8mmHg.
Overpressure protection: double protection for software and hardware.
Measurement range: 10mmHg~270mmHg.
|Continuous real-time measuring.|
Adopt digital SpO2 technology, which has strong anti-interference capability.
Real-time display SpO2, pulse rate, SpO2 Plethysmogram, bar graph, SpO2 and Pulse rate (PR) trend graph.
|Trend graph with the function of adaptive scaling|
|1.SpO2 Measuring||1.Range: 0%~100% (Resolution: 1%)|
2.Accuracy: ±2% in stage of 70%~100%, and unspecified in stage of less than 70%
|Pulse Rate Measuring||1.Range: 30bpm~250bpm(Resolution: 1bpm)|
2.Accuracy: ±2bpm or ±2%, whichever is greater
|Measurement Performance in Weak Filling Condition||SpO2 and pulse rate can be shown correctly when pulse-filling ratio is 0.4%, SpO2 error is ±4%; the Pulse rate error is ±2bpm or ±2%, whichever is greater|
|Resistance to surrounding light||The deviation between the value measured in the condition of man-made light or indoor natural light and that of darkroom is less than ±1%|
|Temperature||5 ˚C~40 ˚C|
|Specific EMC, climate, mechanical environment: Do not use a mobile phone in the vicinity of the veterinary monitor for the strong radiation field produced by mobile phone would interfere with the normal function of the veterinary monitor use.|