Feature: Push button type; Rotation Speed:5~2000RPM fiber optical, match up with LED electric motor special design of the surface,dustproof,safer and more hygeian Sterilization Temperature:135°C Applicable Bur:Φ2.345-2.355mm Connector:E type connector,20:1 implant dental handpiece,match up with Switerland,Japan,Korea and China implant dental system
Operation Instruction: It shall be sterilized before handpiece is used at the first time;sterilization temperature is 135ºC and last for 12 minutes. After using,Please follow the instruction to sterilize the handpiece: 1 extract the bur from the handpiece,use the soft brush to clean the dirt and wash with disinfection of alcohol cotton. 2 lubricate the handpiece,pour two drops into air tube one time. 3 wrap it with sterilization package and mark.
Please do not use the acid solution to scrub, inject or soak the handpiece. Please do not autoclave the handpiece with residual chemical substances. Handpiece should be packed well and place at clean and dry room without dust, sulfur and salt. Please do not touch the handpiece immediately after autoclaving.