Features: Easy removal and easy assembly Dustproof nose design forward & reverse control to adjust control latch tpe, chuck bur firmly.
Instruction: After each treatment,Sterilizing handpiece in autoclave according to the following steps: 1.Remove the bur and brush clean handpiece, clean outside of handpiece with an alcohol solution. 2.Lubricate handpiece, each times drop two drops inside of air inlet pipe. 3.Place into autoclaving pouch and mark. 4.Autoclave the handpiece for 12 min. at 135ºC.
Operation Instruction: It shall be sterilized before handpiece is used at the first time;sterilization temperature is 135ºC and last for 12 minutes. After using,Please follow the instruction to sterilize the handpiece: 1 extract the bur from the handpiece,use the soft brush to clean the dirt and wash with disinfection of alcohol cotton. 2 lubricate the handpiece,pour two drops into air tube one time. 3 wrap it with sterilization package and mark.
Please do not use the acid solution to scrub, inject or soak the handpiece. Please do not autoclave the handpiece with residual chemical substances. Handpiece should be packed well and place at clean and dry room without dust, sulfur and salt. Please do not touch the handpiece immediately after autoclaving.