Business Type: Manufacturer/Factory
Ningbo, Zhejiang, China
Plastic tube flow meter with alarm limit switch
Variable Area Flowmeter Alarm Limit Switch has been designed for use in level alarm applications. Include magnetron and dry reed. It consists of a pair of contacts on ferrous metal reeds in a hermetically sealed glass envelope. The contacts may be normally open, closing when a magnetic field is present, or normally closed and opening when a magnetic field is applied.
2. Application and Features
BJ01: High Alarm Limit (HA)
BJ02: Low Alarm Limit (LA)
Both BJ01 and BJ02 are able to send relevant signals at max, Mid and Min flow,the switches are mounted on certain selected position of rail and whenever magnetic float reaches or passes the switch level,signals will be sent out for internal contact actuation of either opening or closing.
3. Technical Data
Operating Voltage: ≤DC24V or AC24V
Operating Current: ≤0.3A
Constant Current (ON & OFF): ≤0.1A
Rupturing Capacity: ≤1A
Operating Temp: 0°c to +55°c
Protection Levels: IP65
Delay (ON & OFF Position): ≤4mm
4. Installation Hints
Keep off form ambient magnetic disturbance during mounting . such as electric motor.
Flow media should be free from ferromagnetic foreign matter
Ensure pipe or tube to be mounted with the flowmeters free from any potential vibration in case switch contact actuates accidentally.
Datasheet of LZM series flow meter with alarm limit switches
Products with this tag will be sent out within 15 days after you place an order. For the specific sending time and delivery date, please consult the supplier.
*If the supplier with the above tags do not give you the according discount or service, you can click "Contact Us" to make a complaint. If what you said is true, our professional team will help you to handle this problem.
If you are using our website on PC, you can find us by this way:
Go down to the bottom of Made-in-China.com > Go to "Help" Section> Contact Us
If you are using our Made-in-China.com App, you can find us by this way:
My Account > Click setting section which is in the top right corner > Contact Us