Business Type: Manufacturer/Factory & Trading Company
Zigong, Sichuan, China
|PTFE Lined Pipe||10, 15, 20, 25, 32, 40, 50, 65, 80, 100, 125, 150, 200, 250, 300, 350, 400,450, 500, 600|
|Rubber Lined Pipe||40, 50, 65, 80, 100, 125, 150, 200, 300, 350, 400, 500, 600, 800, 1000, 1200,note: special specifications can be customized|
|Flow Direction||positive and negative, net flows Turndown|
|Measuring Range Ratio||150:01:00|
|Accuracy:||piped: 0.5 class, 1.0 class|
|Measured Medium Temperature||Common rubber lining: -20~+60 ° c|
|High temperature rubber lining: -20~+90 ° c|
|PTFE lining: -30~+100 ° c|
|High temperature vinyl lining: -30~+180 ° c|
|Rated Working Pressure||pipeline: DN10~DN65: ≤ 2.5MPa;DN80~DN150: ≤ 1.6MPa;|
|DN200~DN1200: ≤ 1.0MPa|
|Flow Measuring Range||flow measuring range corresponding to flow rate range is 0.3~15m/s|
|Measured Fluid Conductivity||5μs/cm (Most with water as the composition of the medium,its conductivity in the range 200~800 Mu s/cm, optional|
Q: The new magnetic flap liquid level gauge can't show liquid level?
A: 1.check whether there is block in by pass pipe 2.check the the density of medium 3.check whether the float ball is stucked by float ball.
Q: The longest length of magnetic flap liquide level gauge?
A: the length is customized,but in order to be convenient for transport,if the length is over 6m,we recommend
Q: If the float ball is installed at the wrong opposite direction,what's the influence?
A: reading level will be higher than true level,usually higher by length of floatball-70mm
Q: What's installation details of magnetic flap liquid level gauge ?
A: 1.before installation,check specification with label plate. 2.There shouldn't magnetic field within 400m
Q: There is problem in output signal 4--20mA?
A: 1.Check working power supply 2.check wiring 3,check load resistance 4.verify measuring range,zero point
Q: what's specification we need to know before purchasing magnetic flap liquid level gauge?
A: medium density,working temperature,working pressure,material,flange center to center distance,flange standard,whether need 4--20mA output signal;wether need level alarm