Business Type: Manufacturer/Factory & Trading Company
Nanjing, Jiangsu, China
High Quality Laboratory Mixer Machine
Lab Mixer is mainly suitable for mixing tests between different components of dry powders, between particles or between powders and granules in the laboratory. It is the ideal model for exploring the best process parameters of material mixing. It is an advanced experimental equipment for mechatronics.
The HLS-100 laboratory mixer is mounted on a removable base. The main components include racks, swing systems, electrical control systems, hoppers and more. The conical mixing hopper is mainly composed of a bucket body, a bucket cover and a clamp, a pressure block, a butterfly valve and the like. The hopper is mounted on the rotary shaft by cantilever fixing and can be rotated 360 degrees. Since the symmetrical axis of the hopper is at an angle with the rotary axis of the rotary arm, different components of the material move in a three-dimensional space in the hopper, which produces a strong flip and high The shear to achieve the best mixing effect. The slewing system is the key system for completing the mixing function. The slewing motor uses a low-speed high-torque geared motor and is a hybrid power unit. An inductive component is mounted on the rotary shaft assembly. When the rotary body is turned to the set position, the sensor mounted on the frame sends a sampling signal, and the signal passes through the PLC system to read the speed and position information of the hopper, and continuously touches The speed and the number of revolutions are displayed, and the stop command is issued according to the set time. The hopper stops rotating and the hopper stops at the optimal discharge position under the action of the brake motor. The electrical control system is mainly divided into two parts: the electrical cabinet and the sensor installed in the rack. The core control device PLC controller and frequency converter are installed in the electrical control cabinet. During the use, the electrical operation is mainly done on the operation panel (touch screen) of the electrical control cabinet, including the printer, touch screen, power switch and emergency stop switch. . The running time and speed can be displayed on the touch screen interface.